An oscilloscope is a type of electronic test instrument that allows observation of constantly varying signal voltages, usually as a two-dimensional plot of one or more signals as a function of time. Other signals (such as sound or vibration) can be converted to voltages and displayed.
evive Mini Oscilloscope Heat Beat Visualisation
evive has a mini oscilloscope to visualize voltage and current measurements. It has a lot of functions like scaling voltage level, time scaling to see the reading in a better way, a trigger to start the measurement at some set value, scanning modes like single or continuous, offset, save readings to SD card and send data to the computer. It uses a very accurate 24-bit dual channel ADC IC for sensing from Analog Devices (ADE7912).
There are two channels for data acquisition namely Probe I/V and Probe V. The Probe I/V can be used to sense currents or voltage as per the Jumper selection. The maximum sensing frequency is about 1kHz. The current sensing has an accuracy of 3mA with a range of -3A to +3A, while in voltage mode, the accuracy is of 3mV with a range of -5V to +5V. The Probe V can be used for higher voltage sensing in a range of -30V to +30V with an accuracy of 10mV.
The option for Mini -Oscilloscope appears on the main menu. The ports present at the bottom right of evive are used for connecting the evive with the circuit in order to use it as an oscilloscope. The three ports present are Channel 1, COM, and Channel 2 from left to right respectively. COM is the reference from where voltage and current is measured.
From channel 1 you can sense voltage between -30V and +30V. Green color signal plotting is for channel 1.
Opening Mini-Oscilloscope Panel and visualizing sin wave signal
From channel 2 you can either sense voltage from -5V to +5V or sense current from -3A to +3A by changing jumper (JP2). The signal from both these channels is plotted and displayed on the screen. Yellow color is for channel 2.
Sensing the current spikes when motor is stopped with evive mini Oscilloscope
The voltage divisions that can be displayed in evive are:
Voltage Scaling for Channel 1
Voltage Scaling for Channel 2
The voltage division for each channel can be selected according to the amplitude of the signal.
The time divisions that can be displayed in evive are:
evive has the 4 trigger modes available:
It is used to give a DC offset i.e. displacing the amplitude from the mean position for both Channel 1 & 2.
Voltage Offset for Channel 1
Voltage Offset for Channel 2
The options like Voltage divisions & Offset adjustment are displayed twice for Channel 1 & Channel 2.